A "Current out of Control" error on a 2XXX series testboard can be the result of a blown fuse, loose screws, or even a blown amplifier...
AMI networks have become common place in our industry. It is important to know how your network devices will respond to planned and unplanned...
When testing disconnect switches using the WECO-300 model I understand that if testing Sensus meters you require model WECO-300HVA. can a standard...
Did you know that there is a small solenoid behind the socket shield that controls the opening and closing of the jaws? If this solenoid has any...
Take a look at the following webinar to help understand and justify barcoding in your meter shop. This webinar goes into great detail on how to use...
Will the WECO-300 work with a probewell unit?
When testing a form 26s meter using 3ph delta abc power, why does the meter stop or sometimes run backwards when testing at the pf point?
Our newest test board to the marketplace is the 4050X test board. This test board has all the features of the 4000 series test board at a very...
Can you please confirm what meter forms work with the WECO-300.
Does Radian still repair old UTEC units? 620 and 622.
Supported videos include:
Please paste your code into the box below: